Thanks for your help, I appreciate it. The test procedure, which is denoted loopback , is to connect the AI0 channel to the AO0 channel. If the channel testing passed without problems, you may end the testing session by setting 0V on the output in the Voltage Output window. Counting on falling edge. There are 5 items available. Selecting physical channel for the task. Read more about the condition.
|Date Added:||28 June 2008|
|File Size:||30.72 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Have one to sell? Depending on your system and noise levels, connecting pin 4 to ground may help. Attempted delivery on or before the guaranteed date will be considered a timely delivery.
Feedback on optoscience from others who bought this item. Output low is below 0. Refer to eBay Return policy for more details. Before you start using the USB, you should attach labels to the analog and digital terminals. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab No additional import charges on delivery Delivery: In general, a task can contain one or more global or dwq virtual channelsand one such virtual channel is based on one 66009 channel having e.
Thanks for your prompt reply. Report item – opens in a new window or tab.
Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure Please enter a valid Dxq Code. Special USB features are described separately.
Use Single ended mode. The most recent settings are saved, and are loaded the next time MAX is opened. You can create a task for analog output in the same way, that 66009, by selecting Create Task in the right-click menu of the USB device as it appears under Devices and Interfacessee Figure 8.
TechTeach: A quick guide to National Instruments USB and USB multifunction I/O devices
The Dq Input window should indicate the same or almost the same voltage as is set ddaq on AO0. This item may be a floor model or store return that has been used. If a global channel is reconfigured, all tasks which involve that global channel are updated accordingly. Learn More – opens in a new window or tab Returns: Here is a video that shows the steps: I have a a linear potentiometer and a daq You may therefore select one of the upper two options.
I want to wire it. The Voltage Output dialog window in the Test Panels window. But I have a question:. Message 8 of Note that information about an item in the window is shown in the sub-window down right see Figure 609 financing available Select PayPal Credit at checkout to have the option to pay over time.
NI USB Support – National Instruments
Digital input DI and digital output DO: Thanks for your help, I appreciate it. Please enter 5 or 9 numbers for the ZIP Code. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
Email to friends Share on Facebook 609 opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.